Acid Bench 73
|Acid Bench 73|
|Chemicals Used||Sulfuric acid (H2SO4)|
Acid Bench 73 is a wet bench used for working with acids, mostly for etching and cleaning of samples that have had metals and/or polymers on them. It is considered a Metals class tool, so wafers with metal are allowed in the bench with some restrictions. It is intended for use with mixed materials and all substrate sizes up to 6 inches with the main focus being on piece parts.
- Update this with announcements as necessary
Acid Bench 73 is for wet processing of with piece parts, 4, and 6 inch samples of mixed materials. This includes III-V and II-VI materials, samples with polymers, and irregularly shaped samples.
- Feedback controlled hotplate
- Rinse tank can be run as either a Quick dump rinse (QDR) or overflow/cascade tank.
- Sink with goose neck
- Automated Bottle Wash
- Aspirator for chemical cleanup
LNF Provided Chemicals
- Acetic Acid (C2H4O2)
- Aluminum Etch
- Buffered HF (BHF)
- Hydrofluoric acid (HF)
- Chromium Etch 1020
- Hydrochloric acid (HCl)
- Hydrogen peroxide (H2O2)
- Gold Etch GE-8111
- Nitric acid (HNO3)
- Phosphoric Acid (H3PO4)
- Sulfuric acid (H2SO4)
- Feedback control hot plate
- Quick dump rinse (QDR)/Cascade tank
- Automatic Bottle Wash
- Sink with goose neck
- Open deck space
|Warning:||You must use the buddy system for all chemical processing. No processing if you are alone in the lab. If the yellow warning lights are on it means that there are 2 or 1 people in the lab.|
Working with chemicals can be very dangerous. Chemicals used at this bench are some of the most hazardous in the lab. To minimize the chance of accident, make sure you always:
- Wear personal protective equipment. Required PPE at this bench is an apron, face shield, and Trionic gloves.
- LNF Glove Policy
- Know the properties and hazards of chemicals you are working with. If you don’t know, find out by reading about them (review Safety Data Sheet (SDS)) or ask staff.
- Plan the details of your process. Before you start working, plan what quantities of chemicals and sample carriers you will use. Plan how you will rinse your sample and what you will do with any waste generated. Make sure you know how the controllers work.
- Do not use this bench without proper training.
- Waste from this bench goes into the LNF’s acid waste neutralization system and then to city sewers.
- No solvents can be used or disposed of here.
- Solutions which contains toxic materials that cannot be removed by neutralization (such as gold etch and chromium etch) must not be disposed of down the drain.
- Piece Part to 6 inch wafers
- Open to all approved substrates/materials
- Some materials may require the use of private beakers
The Acid Bench 73 is designated as a Metals class tool. Below is a list of approved materials for the tool. Approved means the material is allowed in the tool under normal circumstances. If a material is not listed, please create a helpdesk ticket or email firstname.lastname@example.org for any material requests or questions.
There are a variety of chemical processes that can be performed on this bench, including:
If you wish to perform any other processes on this bench, they must be approved by a tool engineer. Please create a helpdesk ticket for assistance.
Standard operating procedure
For a reader friendly version of the SOP see here For a document version of the SOP, see below Widget text will go here.
- Take the General Wet Bench Training, if you haven't already.
- Read through this page and the Standard Operating Procedure above.
- Practice with your mentor or another authorized user until you are comfortable with tool operation.
- Fill out Process Information Form. You will need to fill this out for each bench you plan on using.
- Open a helpdesk ticket and request a checkout.
- A tool engineer will schedule a time for checkout.
- Complete the 1440 Acid Bench quiz.
- If you pass the quiz the tool engineer will authorize you on the tool. If not you will be contacted to repeat.
- If you break a nitrogen gun or a DI water gun, go in the back service aisle to turn off the inlet to the bench
- Contact staff via Helpdesk system
There is a maintained beaker for gold and Chromium etch - changed every 6 weeks