|Material Processed||Si, Mo and Ge|
|Sample Size||Pieces to 150 mm|
|Supported Processes||Supported Processes|
The Xactix Xetch X.3.B is a xenon difluoride etching tool that will etch silicon, molybdenum, and germanium isotropically. XeF2 is a solid at atmosphere but will sublimate at vacuum. The tool uses two expansion chambers to alternatively sublimate the XeF2 and release it into the etch chamber. The etch rate is highly dependent on the chamber's loading. A wafer with only a thin trench exposed will etch very fast where as a blanket 100 mm Si wafer may etch as slow as 800 Å/cycle. Wafers are manually loaded directly into the chamber, and the tool can accommodate pieces up to 150 mm wafers.
- [2015-06-08] - Added pump bypass to allow for slower chamber pump down.
- Etches Silicon, Molybdenum and Germanium
- High selectivity to many materials including:
- Long undercuts are possible (> 100 µm)
- No stiction associated with wet etching
- Doesn’t quickly attack BOSCH process passivation layer
- Additional Info
- 0 - 6 Torr XeF2
- 0 - 20 Torr N2
- Pieces to 150 mm
- No mounting required
The Xactix XeF2 is designated as a Semi-Clean class tool. Below is a list of approved materials for the tool. Approved means the material is allowed in the tool under normal circumstances. If a material is not listed, please create a helpdesk ticket or email email@example.com for any material requests or questions.
A table of etch rates for the standard process can be seen below, these were all run with 20 second cycle times and 3 Torr XeF2 pressure. More process information for this tool can be found on it's supported processes
|Material||Size||% Open Area||Etch Rate (Å/cycle)|
|SPR 220||100 mm||100%||< 5|
Standard Operating Procedure
Widget text will go here.
- Read through this page and the Standard Operating Procedure above.
- Complete the SOP quiz here.
- Create a helpdesk ticket requesting training.
- A tool engineer will schedule a time for initial training.
- Schedule a checkout session with a tool engineer via the helpdesk ticket system. If this checkout is successful, the engineer will authorize you on the tool.
For a successful checkout session, the user must demonstrate operation of the tool in adherence with this SOP without assistance from the tool owner. We encourage users to use the SOP during the checkout as a guide.
Every other week the chambers leak rate is checked along with the XeF2 usage.